PDS_VERSION_ID = PDS3
RECORD_TYPE = STREAM
OBJECT = TEXT
PUBLICATION_DATE = 20200401
DESCRIPTION = "
This file is intended as documentation of the Field(s) Of View (FOV(s))
for the detectors and/or slits and/or apertures comprising the instrument
on the New Horizons (NH) spacecraft that generated the data archived in
this data set.
This file is a NH Project ALICE SPICE Instrument Kernel (IK),
current at the time of delivery of this data set, with an attached PDS
label prepended. It is only provided as a convenience to the user
to visualize the FOVs of the instrument. This file will not be updated
in this PDS data set as part of any SPICE kernel updates, and should
therefore not be used as a SPICE kernel in any scientific investigation.
Specifically, the references in the IK are not relevant to the graphic
visualization of the FOV and will not be provided with this data set or
archived elsewhere; therefore the references should be ignored in the
context of the intended scope of this file as described above.
As a SPICE IK, this file has much more information than just the
FOV description (e.g. references to project documentation), but in the
context of this PDS data set only the FOV description is relevant. For
a more complete understanding of the geometry and timing issues of the
New Horizons mission, the user is directed to the SPICE PDS data set
for the mission, with a data set ID of NHJ/P/SSSPICE6V1.0.
See further caveats in the PDS NOTE field of this document.
"
NOTE = "
See also the PDS DESCRIPTION field of this document.
CAVEATS:
This file is the NH ALICE SPICE Instrument Kernel (IK),
current at the time of delivery of this data set, with an attached PDS
label prepended. It is only provided as a convenience to the user
to visualize the FOVs of the instrument. This file will not be updated
in this PDS data set as part of any SPICE kernel updates, and should
therefore not be used as a SPICE kernel in any scientific investigation.
Specifically, the references in the IK are not relevant to the graphic
visualization of the FOV and will not be provided with this data set;
therefore the references should be ignored in the context of this file.
If the user wishes to do any data analysis requiring NAIF/SPICE IKs,
they should not use this file, but rather get the most recent IK from
the NH SPICE data set and use that.
 This file is included in the /DOCUMENT/ directory of most if not
all volumes for this instrument as a convenience to the user
because, in some of its sections, it documents the geometry of the
ALICE instrument Field(s) Of View (FOV(s)). Other sections of
this IK (e.g. the references) will have limited use in that scope.
 The original name of the source of this file was
NH_ALICE_V###.TI
where ### is a version number.
 The format of this file, starting five lines after this
TEXT OBJECT, is a SPICE Kernel Pool text file
 The Instrument Kernel itself is (or will be) formally archived
with the New Horizons SPICE dataset.
 See the SPICE documentation for details of that format
 http://naif.jpl.nasa.gov/
 Even without understanding that format, the Instrument Kernel,
and especially its comments, are human readable. Comments are
any line for which one of the following three statements is true:
1) The line is before the first data marker line in the file
2) The line is in a section of lines between a text marker line and
a data marker line with no intervening text or data marker lines
3) The line is in a section of lines between the last text marker and
the end of the file with no intervening text or data marker lines
 a data marker line has the single token '\begindata' on it with
all other characters on the line being whitespace
 a text marker line has the single token '\begintext' on it with
all other characters on the line being whitespace
 N.B. Because padding and a carriage return have been added to
each line of this file, it may or may not be functional
as a valid SPICE kernel.
"
END_OBJECT = TEXT
END
########################################################################
##################### SPICE IK Starts after next line ##################
########################################################################
KPL/IK
ALICE Instrument Kernel
==============================================================================
This instrument kernel (Ikernel) contains references to the mounting
alignment, internal and FOV geometry for the New Horizons ALICE
UV imaging spectroscopy remote sensing package.
Version and Date

The TEXT_KERNEL_ID stores version information of loaded project text
kernels. Each entry associated with the keyword is a string that consists
of four parts: the kernel name, version, entry date, and type. For example,
the ALICE Ikernel might have an entry as follows:
TEXT_KERNEL_ID += 'NEWHORIZONS_ALICE V1.0.0 22FEBRUARY2007 IK'
   
   
KERNEL NAME <+   
  V
VERSION <+  KERNEL TYPE

V
ENTRY DATE
ALICE IKernel Version:
\begindata
TEXT_KERNEL_ID += 'NEWHORIZONS_ALICE V2.0.0 13APR2015 IK'
NAIF_BODY_NAME += ( 'NH_ALICE_SOC' )
NAIF_BODY_CODE += ( 98100 )
NAIF_BODY_NAME += ( 'NH_ALICE_AIRGLOW' )
NAIF_BODY_CODE += ( 98101 )
\begintext
Version 2.0.0  April 13, 2015  Andrew Steffl and Lillian Nguyen
 Described the curvature of the airglow and SOC fields of view
by defining additional boundary corner vectors.
 Added detector row center vectors for SOC and airglow.
 Made the SOC boundary corner vector labeling consistent
with airglow.
 Text updates.
Version 1.2.0  October 16, 2012  Lillian Nguyen
 Updated the airglow field of view.
Version 1.1.1  January 15, 2009  Lillian Nguyen
 Corrected typos in the text.
Version 1.1.0  May 21, 2008  Lillian Nguyen
 Added keywords to describe optical parameters and detector
parameters, and updated the diagrams.
Version 1.0.1  April 11, 2007  Lillian Nguyen
 Corrected the airglow field of view.
Version 1.0.0  February 22, 2007  Lillian Nguyen
 Removed an incorrect annotation from the Alice Slit Design
diagram and updated the remaining diagrams.
 Clarified that the entire Alice slit is visible through
both Alice apertures and updated the field of view
definitions appropriately.
 Noted that the standard acronym for the Solar Occultation
Channel is SOCC.
 Promoting to version 1.0.0 denoting approval of kernel set
by instrument teams.
Version 0.0.2  October 4, 2006  Lillian Nguyen
 Removed the 3letter frame NH_ALI and the Alice base frame.
 Renamed the SOC and Airglow instrument IDs to match the
IDs in the frames kernel.
Version 0.0.1  April 4, 2006  Lillian Nguyen
 Alice SOC and Airglow fields of view redefined according to
orientation diagram received from instrument team.
Version 0.0.0  December 1, 2005  Lillian Nguyen
 Draft Version. NOT YET APPROVED BY INSTRUMENT TEAM.
References

1. ALICE Instrument Specification, 05310.02ISPEC01.
2. ``Kernel Pool Required Reading''
3. Spacecraft to ALICE Interface Control Document (ICD),
73999046.
4. APL New Horizons web site,
http://pluto.jhuapl.edu/spacecraft/overview.html.
5. New Horizons Spacecraft Frames Kernel.
6. New Horizons Mission Science Definitions (MSD),
NH73999000v1.6.
7. New Horizons SOC to Instrument Pipeline ICD,
0531SOCINST01 Rev 0 Chg 0.
8. PALICE_Orientation_on_SC, received from Joel Parker in an
email dated Jan. 25, 2006; discussions with Dave Slater on
Mar. 16 and 23, 2006, and email exchange with Dave Slater on
Mar. 2829, 2006 regarding the diagram.
9. Email from Maarten Versteeg clarifying that the entire
Alice slit is visible through both Alice apertures, received
2/22/2007, and from Joel Parker confirming that we should
change the Alice fields of view to the entire lollipopshaped
slit, received 2/28/2007.
10. Telephone conversation with Hal Weaver about the Alice
instrument.
11. Email from Andrew Steffl regarding Alice pointing offsets,
received on 2/13/2007 and 3/22/2007.
12. Email from Andrew Steffl containing optical and detector
parameters and detector layout information, received on
3/12/2008 and 3/26/2008, and discussion regarding diagrams.
13. Email from Andrew Steffl regarding the Alice airglow field
of view, received on 9/28/2012.
14. Emails from Andrew Steffl received 5/7/2013 and 3/11/2015 
3/18/2015 regarding the additional boundary corner vectors,
slit curvatures, and detector row center vectors for both
Airglow and SOC.
Contact Information

Lillian Nguyen, JHU/APL, (443)7785477, Lillian.Nguyen@jhuapl.edu
Implementation Notes

This file is used by the SPICE system as follows: programs that make use of
this instrument kernel must ``load'' the kernel, normally during program
initialization. Loading the kernel associates data items with their names
in a data structure called the ``kernel pool''. The SPICELIB routine
FURNSH,
CSPICE routine furnsh_c, and IDL routine cspice_furnsh load SPICE kernels
as shown below:
FORTRAN (SPICELIB)
CALL FURNSH ( 'kernel_name' )
C (CSPICE)
furnsh_c ( "kernel_name" )
ICY (IDL)
cspice_furnsh, 'kernel_name'
In order for a program or subroutine to extract data from the pool, the
SPICELIB routines GDPOOL, GCPOOL, and GIPOOL are used. See [2] for details.
This file was created and may be updated with a text editor or word
processor.
Naming Conventions

All names referencing values in this Ikernel start with the characters
`INS' followed by the NAIF New Horizons spacecraft ID number (98)
followed by a NAIF three digit ID code for the ALICE instrument.
The remainder of the name is an underscore character followed by the unique
name of the data item. For example, the airglow boresight direction in the
airglow frame (``NH_ALICE_AIRGLOW''  see [5] ) is specified by:
INS98101_BORESIGHT
The upper bound on the length of the name of any data item is 32
characters.
If the same item is included in more than one file, or if the same item
appears more than once within a single file, the latest value supersedes
any earlier values.
ALICE description

From [4]:
``Alice is an ultraviolet imaging spectrometer that will probe the
atmospheric composition of Pluto. A "spectrometer" is an instrument that
separates light into its constituent wavelengths, like a prism, only
better. An "imaging spectrometer" both separates the different
wavelengths of light and produces an image of the target at each
wavelength.
Alice has two modes of operation: an "airglow" mode, which allows
measurement of emissions from atmospheric constituents, and an
"occultation" mode, when either the Sun or a bright star is viewed
through the atmosphere producing absorption by the atmospheric
constituents. The Alice occultation mode will be used just after New
Horizons passes behind Pluto and looks back at the Sun through Pluto's
atmosphere.''
From [1]:
``ALICE consists of a dedicated telescope that feeds a 0.15m Rowland
circle spectrograph with a spectral passband that spans the extreme and
far ultraviolet (EUV/FUV) wavelength region of 5201870 angstrom. Two
separate input channels, the airglow and the solar occultation channels,
direct light to the telescope. The airglow channel has an input aperture
40 mm x 40 mm with a boresight parallel to the RALPH boresight. The solar
occultation channel (SOC) has a small 0.9 mm diameter aperture located
on the telescope side of the ALICE housing with a boresight near
parallel to the NH spacecraft's High Gain Antenna. A flat relay mirror
at the center of the airglow channel directs the light entering the SOC
aperture to the primary telescope mirror. The reduced SOC aperture
limits the solar UV flux entering the telescope to prevent detector
saturation at the spectrograph focal plane during solar occultation
measurements. A microchannel plate (MCP) doubledelay line (DDL)
detector with dual, solar blind, UVsensitive photocathodes (KBr/CsI),
makes up the instrument's focal plane.
.
.
.
The detector electronics receive detected event pulses from the
detector and provide a digital indication of the spatial and spectral
location of each event being processed. The event processing
electronics receive individual events and process them in one of two
operational modes, pixellist or histogram, based on commands from the
IEM [Integrated Electronics Module]. Histogram mode is used to generate
a two dimensional map of the pixel events with the magnitude of a
particular pixel location indicating the number of times that pixel has
been stimulated. Pixellist mode is used to generate a time ordered
record of detected events. Special periodic data markers (i.e. time
hacks) are inserted in the list of pixels to provide a fixed time
reference for the events. Pixel lists are accumulated continuously in a
"pingpong" memory with the data in one memory bank being transmitted to
the IEM over the science telemetry channel while the other memory is
used to continue data acquisition. The last set of pixel list data and
the complete histogram data are transmitted to the IEM upon the command
to stop data acquisition.
.
.
.
Field of View
Airglow Mode
The airglow slit opening shall have a FOV of 0.1 deg +/ 0.01 deg in the
spectral dimension (slit width) by 4 deg +/ 0.1 deg in the spatial
dimension (slit length). The center of the airglow slit shall be offset
from the ALICE optical boresight by 1 +/ 0.1 deg.
SOC Mode
The SOC slit opening shall have a FOV of 2 deg +/ 0.1 deg in the
spectral dimension (slit width) by 2 deg +/ 0.1 deg in the spatial
dimension (slit length). The center of the SOC slit shall be offset from
the ALICE optical boresight by 2 +/ 0.1 deg, and shall be a contiguous
opening with the airglow slit.
The boresight of the SOC FOV shall be aligned to the HGA (i.e. REX)
boresight as specified in the Alice ICD (73999046).
The coalignment error between the ALICE SOC boresight and the REX
boresight shall be as specified in the Alice ICD (73999046).
The SOC aperture shall be sized to limit the detector output count
rate to less than 30 kHz during the PlutoCharon solar occultation.
.
.
.
ORIENTATION
AIRGLOW CHANNEL
APL shall align the ALICE Airglow channel to the identical
requirements as RALPH/MVIC [25] section 6.5.2.2.1; namely:
a. The boresight shall be aligned to the spacecraft X axis to within
0.90 deg.
b. The direction perpendicular to the 1024element lines, and lying in
the focal plane, shall be aligned at 90 deg +/ 0.90 to the scan
axis of the spacecraft. {11/19/02}
SOLAR OCCULTATION CHANNEL
APL shall align the solar occultation channel (SOC) boresight to lie
within 0.5 deg of a line in the YZ plane righthand rotated by 2 deg
around the X axis from the REX boresight. {11/19/02}''
From [3]:
``PERSI
PERSI [Pluto Exploration Remote Sensing Investigation] consists of two
separate instruments mounted close together on the New Horizons
spacecraft:
A. RALPH, the name (not an acronym) for the instrument consisting of:
(1) MVIC, the Multispectral Visible Imaging Camera, including focal
plane, electronics, telescope and mechanical structure, and
(2) LEISA, the Linear Etalon Imaging Spectral Array, a focal plane
assembly inside the MVIC structure that performs as an infrared
(IR) mapping spectrometer.
B. ALICE, the name (not an acronym) for PERSI's ultraviolet mapping
spectrometer {8/8/02}
.
.
.
ALICE will be mounted directly to the +Z side of the spacecraft''
ALICE Field of View Parameters

The Alice instrument has two apertures, the Solar Occulation Channel (SOC),
and the Airglow aperture. The SOC is now generally referred to as SOCC
(Solar OCcultation Channel) to distinguish it from the SOC (Science
Operations Center). Since the documents to which this kernel refers still
use the SOC acronym for the Solar Occultation Channel, we continue to use
it here as well, although SOCC is becoming the standard term [9].
Light travels through a lollipopshaped slit described below for both the
SOC and airglow apertures. Although [3], Rev A, defines the SOC field of
view as only the 2.0 x 2.0 degree "box" portion of the lollipopshaped
slit and the Airglow field of view as the 0.1 x 4.0 degree "stem" of the
lollipopshaped slit, the complete lollipop shape is visible through both
apertures [9].
The spectral resolution of airglow is degraded in the 2.0 x 2.0 degree
"box" portion of the slit, but data is nevertheless still available in that
portion of the slit. The 2.0 x 2.0 degree "box" portion of the slit is
wider than the narrow "stem" portion to observe the sun during Solar
Occulation Channel operations, but it is also possible for the sun to
appear in the narrower "stem" portion [10]. Hence, the fields of view for
both apertures have been defined here to be the entire lollipopshaped
slit even though they are defined differently in [3], Rev A.
The diagram below (reproduced from Figure 19 of [3], and [12]) illustrates
the projection of the ALICE field of view onto the sky. The ALICE entrance
slit design shows the 2.0 x 2.0 degree "box" portion of the lollipopshaped
slit and the 0.1 x 4.0 degree "stem" portion of the lollipopshaped slit.
This diagram shows the actual position of the Alice slit on the spacecraft,
using the spacecraft coordinate system. It is important to note that the
entire 'lollipop' shape of the Alice slit is visible through both the SOC
and the airglow apertures [9]. The positions of detector rows 6 and 25 are
shown in the diagram below [12]. Rows 15 and 2632 are effectively
masked. The slit does not image onto those detector rows [14].
ALICE SLIT DESIGN
_____________________4_________________________
  
  
 .3. row 25 
    
^     
Z    2   
(deg)      
     
  '1' 
    
    
+++ +++
4 3 2 1   1 2 3 4
   
 1  
   
   
 2  
   
   
 3'' row 6 
  
  
____________________4________________________
>
+Y into the page for SOC X (deg) for SOC
X into the page for airglow Y (deg) for airglow
Note that because of the 2.0 degree instrument tip to rotate the +Y axis
(the REX antenna boresight) to the center of the box part of the Alice slit
when using the SOC field of view [8], the axis labeling for the airglow
aperture in the diagram above is slightly incorrect. The horizontal axis
for airglow is not exactly +Y, as it is labeled, but actually 2.0 degrees
rotated about spacecraft +X from spacecraft +Y. The same is true of the
vertical axis for airglow.
SOC Channel FOV Definition
In SOC instrument coordinates, the boresight is the +Y axis, and the
X and Z axes are close to the spacecraft X and Z axes, respectively [8].
A discussion of the SOC coordinate frame relative to the spacecraft frame
can be found in [5]. The boresight of the SOC field of view is in the
center of the 2.0 x 2.0 degree 'box' portion of the lollipop [9].
We use a polygon to describe the lollipopshaped field of view, and will
determine twentyeight boundary corner vectors defining the vertices of
the polygon. The SOC coordinate system, field of view center, and
selected vertices of the polygonal field of view are illustrated below.
The vertices labeled in the diagram appear to be bounded by straight
edges, but the SOC field of view has a slight curvature, described
later. The vertices not shown are equally spaced in increments of 0.5
degree along the instrument Z axis between those shown here.
SOC Instrument
___ V0 _ V27 Coordinate System
^   row 6
   ^ +Z
    inst
   
   
   
   x>
   +Y (in) +X
   inst inst
  
 6 deg  
  
  
  
  
 ___ V9_____ _____V18
 ^  V8 V19 
   
   
  2 deg  x  < SOC boresight 
    ^
     1 deg
_v_ _v_ _____________ _v_
V13 V14 row 25
<>
2 deg
The two diagrams below illustrate the calculations used to determine the
X and Z components of the boundary corner vectors shown above, in the
case of no curvature. The Y component of each of the boundary corner
vectors is arbitrarily set to 1 to simplify the calculations (i.e. for
ease of computation, the field of view vectors are terminated at the
plane Y = 1).
View looking down the SOC X axis

^ Z
 inst


z0, z27 oo V0, V27 
 / 
  
 /  
  
 /  
  
 /  
  
 /  
  
 /  
   4.0 deg
 /  
  
 /  
  
 /  
  
 /  
  
 /  
  
 /  
z8, z9, z18, z19 .o V8, V9, V18, V19 
 / .' 
 .'  
 / .'   1.0 deg
 .'  
/'  y=1 
X (out) o+> Y (boresight) 
inst `.  inst 
 `.  
 `.   1.0 deg
 `.  
 `. 
z13, z14 oo V13, V14 



v
View looking up the SOC Z axis

^ X
 inst


x14x18 oo V14V18 
 .' 
 .'  
 .'   0.95 deg
 .'  
 .'  
x19x27 o .' ___...o V19V27 
 .'.'''  y=1  0.05 deg
Z (in) x+> Y (boresight) 
inst  `.`...___  inst  0.05 deg
x0x8 o `. ```o V0V8 
 `.  
 `.  
 `.   0.95 deg
 `.  
 `. 
x9x13 oo V9V13 


v
If there is no curvature to the field of view, the upper diagram
gives the Z coordinates of the field of vectors (in degrees) as:
tan(5.0) = z0 = z27
tan(1.0) = z8 = z9 = z18 = z19
tan(1.0) = z13 = z14
And the lower diagram gives X coordinates (in degrees) as:
tan(1.0) = x14 through x18
tan(0.05) = x19 through x27
tan(0.05) = x0 through x8
tan(1.0) = x9 through x13
Which yields the following boundary corner vectors:
V0 = [x0, y, z0] = [tan(0.05), 1, tan( 5.0)]
V8 = [x8, y, z8] = [tan(0.05), 1, tan( 1.0)]
V9 = [x9, y, z9] = [tan(1.0 ), 1, tan( 1.0)]
V13 = [x13, y, z13] = [tan(1.0 ), 1, tan(1.0)]
V14 = [x14, y, z14] = [tan( 1.0 ), 1, tan(1.0)]
V18 = [x18, y, z18] = [tan( 1.0 ), 1, tan( 1.0)]
V19 = [x19, y, z19] = [tan( 0.05), 1, tan( 1.0)]
V27 = [x27, y, z27] = [tan( 0.05), 1, tan( 5.0)]
From [14]:
Note that the projection of the slit onto the sky is not actually
rectilinear, but rather is somewhat curved. This is due to optical
distortion (mostly coma) introduced by the primary mirror. We fit the
observed X offset of the middle of the slit as a function of offset in
the Z dimension using a 2nd degree polynomial of the form:
X_off(Z) = a + b * Z + c * Z^2
where X and Z are the offsets from the instrument boresight, measured in
degrees. The best fit values for the polynomial coefficients are:
a = 0.011711347879220315
b = 0.021970015575058958
c = 0.004223195198431527
The actual boundary corner vectors for the SOC slit are then
V0 = [tan(X_off( 5.0)  0.05), 1, tan( 5.0)];
V1 = [tan(X_off( 4.5)  0.05), 1, tan( 4.5)];
V2 = [tan(X_off( 4.0)  0.05), 1, tan( 4.0)];
V3 = [tan(X_off( 3.5)  0.05), 1, tan( 3.5)];
V4 = [tan(X_off( 3.0)  0.05), 1, tan( 3.0)];
V5 = [tan(X_off( 2.5)  0.05), 1, tan( 2.5)];
V6 = [tan(X_off( 2.0)  0.05), 1, tan( 2.0)];
V7 = [tan(X_off( 1.5)  0.05), 1, tan( 1.5)];
V8 = [tan(X_off( 1.0)  0.05), 1, tan( 1.0)];
V9 = [tan(X_off( 1.0)  1.0 ), 1, tan( 1.0)];
V10 = [tan(X_off( 0.5)  1.0 ), 1, tan( 0.5)];
V11 = [tan(X_off( 0.0)  1.0 ), 1, tan( 0.0)];
V12 = [tan(X_off(0.5)  1.0 ), 1, tan(0.5)];
V13 = [tan(X_off(1.0)  1.0 ), 1, tan(1.0)];
V14 = [tan(X_off(1.0) + 1.0 ), 1, tan(1.0)];
V15 = [tan(X_off(0.5) + 1.0 ), 1, tan(0.5)];
V16 = [tan(X_off( 0.0) + 1.0 ), 1, tan( 0.0)];
V17 = [tan(X_off( 0.5) + 1.0 ), 1, tan( 0.5)];
V18 = [tan(X_off( 1.0) + 1.0 ), 1, tan( 1.0)];
V19 = [tan(X_off( 1.0) + 0.05), 1, tan( 1.0)];
V20 = [tan(X_off( 1.5) + 0.05), 1, tan( 1.5)];
V21 = [tan(X_off( 2.0) + 0.05), 1, tan( 2.0)];
V22 = [tan(X_off( 2.5) + 0.05), 1, tan( 2.5)];
V23 = [tan(X_off( 3.0) + 0.05), 1, tan( 3.0)];
V24 = [tan(X_off( 3.5) + 0.05), 1, tan( 3.5)];
V25 = [tan(X_off( 4.0) + 0.05), 1, tan( 4.0)];
V26 = [tan(X_off( 4.5) + 0.05), 1, tan( 4.5)];
V27 = [tan(X_off( 5.0) + 0.05), 1, tan( 5.0)];
These vectors are given in the field of view definition below,
starting with V0.
\begindata
INS98100_FOV_FRAME = 'NH_ALICE_SOC'
INS98100_FOV_SHAPE = 'POLYGON'
INS98100_BORESIGHT = ( 0.0, 1.0, 0.0 )
INS98100_FOV_CLASS_SPEC = 'CORNERS'
INS98100_FOV_BOUNDARY_CORNERS = (
0.00100253752889 1.00000000000000 0.08748866352592
0.00084414580723 1.00000000000000 0.07870170682462
0.00072260847678 1.00000000000000 0.06992681194351
0.00063792551242 1.00000000000000 0.06116262015048
0.00059009689984 1.00000000000000 0.05240777928304
0.00057912263212 1.00000000000000 0.04366094290851
0.00060500270782 1.00000000000000 0.03492076949175
0.00066773713045 1.00000000000000 0.02618592156919
0.00076732590960 1.00000000000000 0.01745506492822
0.01734969416001 1.00000000000000 0.01745506492822
0.01748617861330 1.00000000000000 0.00872686779076
0.01765952951783 1.00000000000000 0.00000000000000
0.01786974755832 1.00000000000000 0.00872686779076
0.01811683358001 1.00000000000000 0.01745506492822
0.01679331155987 1.00000000000000 0.01745506492822
0.01704038829940 1.00000000000000 0.00872686779076
0.01725060179759 1.00000000000000 0.00000000000000
0.01742395127692 1.00000000000000 0.00872686779076
0.01756043608392 1.00000000000000 0.01745506492822
0.00097800380481 1.00000000000000 0.01745506492822
0.00107759263788 1.00000000000000 0.02618592156919
0.00114032711226 1.00000000000000 0.03492076949175
0.00116620721331 1.00000000000000 0.04366094290851
0.00115523293456 1.00000000000000 0.05240777928304
0.00110740427880 1.00000000000000 0.06116262015048
0.00102272125756 1.00000000000000 0.06992681194351
0.00090118388923 1.00000000000000 0.07870170682462
0.00074279219559 1.00000000000000 0.08748866352592
)
\begintext
The following keyword defines the look vectors in instrument coordinates
corresponding to the center of each row of the detector, starting with
row 1 [14].
\begindata
INS98100_ROW_CENTERS = (
0.00118488759652 1.00000000000000 0.10938797626329
0.00097308343851 1.00000000000000 0.10547311580785
0.00077744179592 1.00000000000000 0.10144595647101
0.00059796263701 1.00000000000000 0.09730665189832
0.00043464593812 1.00000000000000 0.09305534087148
0.00028749168221 1.00000000000000 0.08869214574564
0.00015649985766 1.00000000000000 0.08421717084747
0.00004167045704 1.00000000000000 0.07963050083353
0.00005699652387 1.00000000000000 0.07493219900812
0.00013950108699 1.00000000000000 0.07012230559955
0.00020584323273 1.00000000000000 0.06520083599366
0.00025602296060 1.00000000000000 0.06016777892343
0.00029004026979 1.00000000000000 0.05502309461292
0.00030789515948 1.00000000000000 0.04976671287400
0.00030958762920 1.00000000000000 0.04439853115374
0.00029511767889 1.00000000000000 0.03891841253046
0.00026448530896 1.00000000000000 0.03332618365580
0.00021769052015 1.00000000000000 0.02762163264025
0.00015473331333 1.00000000000000 0.02180450687909
0.00007561368907 1.00000000000000 0.01587451081549
0.00001966835279 1.00000000000000 0.00983130363712
0.00013111281379 1.00000000000000 0.00367449690240
0.00025871969761 1.00000000000000 0.00259634790791
0.00040248901091 1.00000000000000 0.00898172191857
0.00056242076437 1.00000000000000 0.01548217147205
0.00073851497376 1.00000000000000 0.02209830083073
0.00093077166126 1.00000000000000 0.02883077497505
0.00113919085679 1.00000000000000 0.03568032250399
0.00136377259954 1.00000000000000 0.04264773864499
0.00160451693960 1.00000000000000 0.04973388838087
0.00186142393970 1.00000000000000 0.05693970970183
0.00213449367714 1.00000000000000 0.06426621699143
)
\begintext
Airglow Channel FOV Definition
In airglow instrument coordinates, the boresight is the X axis, and the
+Y and +Z axes in the instrument frame are close to the spacecraft +Y and
+Z axes, respectively [8]. A discussion of the airglow coordinate frame
relative to the spacecraft frame can be found in [5].
We use a polygon to describe the lollipopshaped field of view, and will
determine twentyeight boundary corner vectors defining the vertices of
the polygon. The airglow coordinate system, field of view center, and
selected vertices of the polygonal field of view are illustrated below.
The vertices labeled in the diagram appear to be bounded by straight
edges, but the airglow field of view has a slight curvature, described
later. The vertices not shown are equally spaced in increments of 0.5
degree along the instrument Z axis between those shown here. Refer to
the SOC calculations for determining the boundary corner vectors in
the case of no curvature. Airglow boundary corner vectors will
terminate at the plane X = 1.
Airglow Instrument
___ V0 _ V27 ___ Coordinate System
^   ^ row 6
    ^ +Z
     inst
    
    
    3 deg 
    x>
    X (in) +Y
    inst inst
   
 6 deg   
   v
 x < Airglow boresight 
   ^
   
 ___ V9_____ _____V18  2.0 degree
difference
 ^  V8 V19   between SOC
boresight
     and airglow
boresight
    v
  2 deg  *  
    ^
     1.0 degree
_v_ _v_ _____________ _v_
V13 V14 row 25
From [14]:
Note that the projection of the slit onto the sky is not actually
rectilinear, but rather is somewhat curved. This is due to optical
distortion (mostly coma) introduced by the primary mirror. We fit the
observed Y offset of the middle of the slit as a function of offset
in the Z dimension using a 2nd degree polynomial of the form:
Y_off(Z) = a + b * Z + c * Z^2
where Y and Z are the offsets from the instrument boresight, measured in
degrees. The best fit values for the polynomial coefficients are:
a = 9.4890826830508E05
b = 0.0045053173995892
c = 0.0055159552582139
The twentyeight airglow boundary corner vectors can be determined using a
similar
calculation as was done for the SOC and by applying the above polynomial to
the Y
coordinates. The calculations for the boundary corner vectors are then:
V0 = [1, tan(Y_off_airglow( 3.0)  0.05), tan( 3.0)];
V1 = [1, tan(Y_off_airglow( 2.5)  0.05), tan( 2.5)];
V2 = [1, tan(Y_off_airglow( 2.0)  0.05), tan( 2.0)];
V3 = [1, tan(Y_off_airglow( 1.5)  0.05), tan( 1.5)];
V4 = [1, tan(Y_off_airglow( 1.0)  0.05), tan( 1.0)];
V5 = [1, tan(Y_off_airglow( 0.5)  0.05), tan( 0.5)];
V6 = [1, tan(Y_off_airglow( 0.0)  0.05), tan( 0.0)];
V7 = [1, tan(Y_off_airglow(0.5)  0.05), tan(0.5)];
V8 = [1, tan(Y_off_airglow(1.0)  0.05), tan(1.0)];
V9 = [1, tan(Y_off_airglow(1.0)  1.0 ), tan(1.0)];
V10 = [1, tan(Y_off_airglow(1.5)  1.0 ), tan(1.5)];
V11 = [1, tan(Y_off_airglow(2.0)  1.0 ), tan(2.0)];
V12 = [1, tan(Y_off_airglow(2.5)  1.0 ), tan(2.5)];
V13 = [1, tan(Y_off_airglow(3.0)  1.0 ), tan(3.0)];
V14 = [1, tan(Y_off_airglow(3.0) + 1.0 ), tan(3.0)];
V15 = [1, tan(Y_off_airglow(2.5) + 1.0 ), tan(2.5)];
V16 = [1, tan(Y_off_airglow(2.0) + 1.0 ), tan(2.0)];
V17 = [1, tan(Y_off_airglow(1.5) + 1.0 ), tan(1.5)];
V18 = [1, tan(Y_off_airglow(1.0) + 1.0 ), tan(1.0)];
V19 = [1, tan(Y_off_airglow(1.0) + 0.05), tan(1.0)];
V20 = [1, tan(Y_off_airglow(0.5) + 0.05), tan(0.5)];
V21 = [1, tan(Y_off_airglow( 0.0) + 0.05), tan( 0.0)];
V22 = [1, tan(Y_off_airglow( 0.5) + 0.05), tan( 0.5)];
V23 = [1, tan(Y_off_airglow( 1.0) + 0.05), tan( 1.0)];
V24 = [1, tan(Y_off_airglow( 1.5) + 0.05), tan( 1.5)];
V25 = [1, tan(Y_off_airglow( 2.0) + 0.05), tan( 2.0)];
V26 = [1, tan(Y_off_airglow( 2.5) + 0.05), tan( 2.5)];
V27 = [1, tan(Y_off_airglow( 3.0) + 0.05), tan( 3.0)];
The twenty eight airglow boundary corner vectors are given in the field of
view
definition below, starting with V0.
\begindata
INS98101_FOV_FRAME = 'NH_ALICE_AIRGLOW'
INS98101_FOV_SHAPE = 'POLYGON'
INS98101_BORESIGHT = ( 1.0, 0.0, 0.0 )
INS98101_FOV_CLASS_SPEC = 'CORNERS'
INS98101_FOV_BOUNDARY_CORNERS = (
1.00000000000000 0.00024377442976 0.05240777928304
1.00000000000000 0.00046920499491 0.04366094290851
1.00000000000000 0.00064649979577 0.03492076949175
1.00000000000000 0.00077565881616 0.02618592156919
1.00000000000000 0.00085668203475 0.01745506492822
1.00000000000000 0.00088956943408 0.00872686779076
1.00000000000000 0.00087432100614 0.00000000000000
1.00000000000000 0.00081093675472 0.00872686779076
1.00000000000000 0.00069941669428 0.01745506492822
1.00000000000000 0.01728176461970 0.01745506492822
1.00000000000000 0.01712206158924 0.02618592156919
1.00000000000000 0.01691420983116 0.03492076949175
1.00000000000000 0.01665821034766 0.04366094290851
1.00000000000000 0.01635406434452 0.05240777928304
1.00000000000000 0.01855610781879 0.05240777928304
1.00000000000000 0.01825194166977 0.04366094290851
1.00000000000000 0.01799593023441 0.03492076949175
1.00000000000000 0.01778807213729 0.02618592156919
1.00000000000000 0.01762836628488 0.01745506492822
1.00000000000000 0.00104591305314 0.01745506492822
1.00000000000000 0.00093439294697 0.00872686779076
1.00000000000000 0.00087100868890 0.00000000000000
1.00000000000000 0.00085576026146 0.00872686779076
1.00000000000000 0.00088864766074 0.01745506492822
1.00000000000000 0.00096967089531 0.02618592156919
1.00000000000000 0.00109882998854 0.03492076949175
1.00000000000000 0.00127612498424 0.04366094290851
1.00000000000000 0.00150155595557 0.05240777928304
)
\begintext
The following keyword defines the look vectors in instrument coordinates
corresponding to the center of each row of the detector, starting with
row 1 [14].
\begindata
INS98101_ROW_CENTERS = (
1.00000000000000 0.00140952012553 0.07446590186039
1.00000000000000 0.00124663824582 0.07049438235957
1.00000000000000 0.00109018756000 0.06643296168342
1.00000000000000 0.00094087103852 0.06228165812733
1.00000000000000 0.00079940760111 0.05804047493042
1.00000000000000 0.00066653211700 0.05370939926237
1.00000000000000 0.00054299540494 0.04928840118661
1.00000000000000 0.00042956423330 0.04477743259898
1.00000000000000 0.00032702131999 0.04017642614118
1.00000000000000 0.00023616533236 0.03548529408805
1.00000000000000 0.00015781088706 0.03070392720762
1.00000000000000 0.00009278854981 0.02583219359293
1.00000000000000 0.00004194483517 0.02086993746435
1.00000000000000 0.00000614220625 0.01581697794109
1.00000000000000 0.00001374092552 0.01067310778044
1.00000000000000 0.00001681020062 0.00543809208322
1.00000000000000 0.00000215531159 0.00011166696369
1.00000000000000 0.00003114999693 0.00530646181792
1.00000000000000 0.00008404792841 0.01081662026217
1.00000000000000 0.00015749663478 0.01641916757870
1.00000000000000 0.00025247021722 0.02211449767833
1.00000000000000 0.00036995872725 0.02790304071145
1.00000000000000 0.00051096816844 0.03378526465536
1.00000000000000 0.00067652049874 0.03976167695345
1.00000000000000 0.00086765363356 0.04583282620936
1.00000000000000 0.00108542144984 0.05199930393935
1.00000000000000 0.00133089379118 0.05826174638642
1.00000000000000 0.00160515647437 0.06462083640010
1.00000000000000 0.00190931129736 0.07107730538585
1.00000000000000 0.00224447604910 0.07763193532855
1.00000000000000 0.00261178452137 0.08428556089465
1.00000000000000 0.00301238652307 0.09103907161807
)
\begintext
ALICE Optics Parameters

ALICE has the following optics parameters:

parameter Airglow SOCC

Focal length (mm) 120 120
fnumber 3 120
IFOV (spatial) (degrees/pixel) 0.3 0.3
Aperture diameter (mm) 40 1

These parameters are captured in the following keywords in the same units
as in the table:
\begindata
INS98100_FOCAL_LENGTH = ( 120 )
INS98100_F/NUMBER = ( 120 )
INS98100_IFOV = ( 0.3 )
INS98100_APERTURE_DIAMETER = ( 1 )
INS98101_FOCAL_LENGTH = ( 120 )
INS98101_F/NUMBER = ( 3 )
INS98101_IFOV = ( 0.3 )
INS98101_APERTURE_DIAMETER = ( 40 )
\begintext
ALICE Detector Parameters

ALICE has the following detector parameters:

parameter

Detector size in pixels 1024 x 32
Detector center (511.5, 16)

These parameters are captured in the following keywords in the same units
as in the table (note that pixel numbers begin at 0):
\begindata
INS98100_PIXEL_SAMPLES = ( 1024 )
INS98100_PIXEL_LINES = ( 32 )
INS98100_DETECTOR_CENTER = ( 511.5, 16 )
INS98101_PIXEL_SAMPLES = ( 1024 )
INS98101_PIXEL_LINES = ( 32 )
INS98101_DETECTOR_CENTER = ( 511.5, 16 )
\begintext